|Item Weight||15.2 ounces|
|Package Dimensions||9.3 x 4.4 x 2.4 inches|
|Item model number||CECOMINOD056574|
|Size||208mm x 78mm x 35mm|
|Power Source||1.5V AAA battery|
|Amperage Capacity||400 A|
|Item Package Quantity||1|
|Number Of Pieces||1|
|Special Features||Square value measuring function. (at AC A and AC V range), Maximum & minimum value measuring function., Converte rate: 3 times/sec., Bar graph: 30 times/sec., Auto power off., Continuity test., Diode test., MAX. diameter for conductor: Ф26mm., Secondary injection molding case., Conform to the IEC1010-1 & IEC1010-1-032 standards.|
|Usage||measuring current, voltage, and resistance|
|Battery Cell Type||Alkaline|
AIMO MS2108A Auto Range Digital Clamp Meter 400 AC DC Current Hz Tester
|Price:||$39.99 & FREE Shipping. Details|
Fulfillment by Amazon (FBA) is a service we offer sellers that lets them store their products in Amazon's fulfillment centers, and we directly pack, ship, and provide customer service for these products. Something we hope you'll especially enjoy: FBA items qualify for FREE Shipping and .
If you're a seller, Fulfillment by Amazon can help you increase your sales. We invite you to learn more about Fulfillment by Amazon .
- Enter your model number to make sure this fits.
- Measures AC/DC amps, AC/DC voltage, resistance, frequency, duty, capacitance, as well as continuity and diode test
- Display: 4000 counts LCD display with analog bar
- Maximum voltage between terminals and earth ground: DC 1000V or AC 750V
- Over range protection for all ranges
- Reading hold function
Frequently bought together
Customers who bought this item also bought
Perform measurements of AC/DC voltage and current, resistance, frequency, duty, capacitance, as well as continuity and diode test.
Display: 4000 counts with analog bar LCD display.
Maximum voltage between terminals and earth ground: DC 1000V or AC 750V.
Over range protection for all ranges.
Reading hold function.
Square value measuring function. (at AC A and AC V range)
Maximum & minimum value measuring function.
Converte rate: 3 times/sec.
Bar graph: 30 times/sec.
Auto power off.
MAX. diameter for conductor: Ф26mm.
Secondary injection molding case.
Conform to the IEC1010-1 & IEC1010-1-032 standards.
Protection class CAT III 600V and pollution degree 2
Display: 4000 counts
DC Voltage Accuracy
400mV / 4V / 40V / 400V / 1000V ± 0.8%
AC Voltage Accuracy
4V / 40V / 400V / 750V ± 1.0%
DC Current Accuracy
40A / 400A ± 2%
AC Current Accuracy
40A / 400A ± 2%
400Ω / 4kΩ / 40kΩ / 400kΩ ± 0.8%, 4MΩ / 40MΩ ± 1.2%
40nF / 400nF / 4μF / 40μF / 400μF / 4000μF ± 4%
10Hz / 100Hz / 1KHz / 10kHz / 100kHz / 1MHz / 10MHz ± 1%, 10 ~10KHz ± 2%
Duty Cycle: 0.1% ~ 99.9% ± 3%
Operating Temperature: 0℃ ~ 40℃ (32°F ~ 104°F)
Storage Temperture: -10℃ ~ 50℃ (10°F ~ 122°F)
MAX. Diameter for Jaw: Ф26mm
Dimension: 208mm x 78mm x 35mm
Power: 1.5V (AAA) x 3 (not included)
1 x MS2108A Clamp Meter
1 x Test Leads
1 x Canvas Bag
Top customer reviews
I am pleasantly surprised. It reads either .5 amps low or is right on. (All meters will disagree a few tenths from each other depending on each's error) (I tend to compare my existing meters with someone else's brand new one for reference.) My twenty year old fluke AC clamp meter now reads .5 amps low thus I compensate for it. I bought this for measuring DC amps when I am running certain appliances off a 12 volt DC to AC 120 volt AC inverter. (I want to know what is coming out of my two golf cart batteries) It was simpler than purchasing a permanent DC voltmeter/ammeter. The leads are decent quality.
I like the basic accuracy, and I like the built in light which allows you to see and clamp around wires in the dark. My basic initial impression is of good quality. The price is great.
It does all what I need , R, Vac, Vdc,Iac
It even possess a DC amp meter which is rare for that price .
The only thing I would have to say is
That the DC amp is difficult to 0 out, it always shows DC current of 0.25 to 0.5 Amp even if no wire passes into the clamp.
Apart from that , this meter is near perfect!
I used it to fine tune my PV install on my RV, and I cleared out all problems